You are here: Specifications > EMVA 1288 Imaging Performance
EMVA 1288 Imaging Performance
Measurements are taken based on guidelines in the EMVA 1288 standard; the full definition can be found at EMVA.org. Camera settings are: maximum bit depth, 16-bit pixel format, and ISP disabled. The center wavelength is 525 nm unless otherwise noted. Results are captured at room temperature (20°C).
|Sensor||IMX297 (1/2.9, CMOS, Global Shutter)|
|ADC Bit Depth||10|
|Quantum Efficiency Mono (% at 525 nm)||64.34|
|Quantum Efficiency Blue (% at 470 nm)||N/A|
|Quantum Efficiency Green (% at 525 nm)||N/A|
|Quantum Efficiency Red (% at 630 nm)||N/A|
|Temporal Dark Noise (Read Noise) (e-)||6.99|
|Temporal Dark Noise (Read Noise) (DN)||0.33|
|Signal to Noise Ratio Maximum (dB)||43.21|
|Signal to Noise Ratio Maximum (Bits)||7.18|
|Absolute Sensitivity Threshold (γ)||11.64|
|Absolute Sensitivity Threshold (e-)||7.49|
|Saturation Capacity (Well Depth) (e-)||20924.80|
|Saturation Capacity (Well Depth) (γ)||32523.10|
|Dynamic Range (dB)||68.93|
|Dynamic Range (Bits)||11.45|
Firefly FFY-U3-04S2M Spectral Response Curve
© 2015-2020 FLIR® Integrated Imaging Solutions Inc.
All rights reserved.
Legal | Contact Support