You are here: Specifications > EMVA 1288 Imaging Performance
EMVA 1288 Imaging Performance
Measurements are taken based on guidelines in the EMVA 1288 standard; the full definition can be found at EMVA.org. Camera settings are: maximum bit depth, 16-bit pixel format, and ISP disabled. The center wavelength is 525 nm unless otherwise noted. Results are captured at room temperature (20°C).
|Sensor||IMX296 (1/2.9, CMOS, Global Shutter)|
|ADC Bit Depth||10|
|Quantum Efficiency Mono (% at 525 nm)||65.45|
|Quantum Efficiency Blue (% at 470 nm)||N/A|
|Quantum Efficiency Green (% at 525 nm)||N/A|
|Quantum Efficiency Red (% at 630 nm)||N/A|
|Temporal Dark Noise (Read Noise) (e-)||4.91|
|Temporal Dark Noise (Read Noise) (DN)||0.45|
|Signal to Noise Ratio Maximum (dB)||40.26|
|Signal to Noise Ratio Maximum (Bits)||6.69|
|Absolute Sensitivity Threshold (γ)||8.26|
|Absolute Sensitivity Threshold (e-)||5.41|
|Saturation Capacity (Well Depth) (e-)||10612.20|
|Saturation Capacity (Well Depth) (γ)||16214.00|
|Dynamic Range (dB)||65.86|
|Dynamic Range (Bits)||10.94|
Firefly FFY-U3-16S2M-DL Spectral Response Curve
© 2015-2020 FLIR® Integrated Imaging Solutions Inc.
All rights reserved.
Legal | Contact Support